Last Updated10th of January, 2012
Indicates calibration error. This can be due to:
-No standard solution on the sensor
-Bubbles on the surface of the ISFET sensor chip
-No connection between the ISFET sensor chip and the liquid junction by means of the standard solution
-Reference electrode not properly set in the pH meter
-Reference electrode service life is over.
-Corrosion on Circuit Board